Ultrafast spoiled gradient echo sequences
|Type of sequence||Philips||Siemens||GE||Hitachi||Toshiba|
|Ultrafast spoiled GE|| |
Ultrafast spoiled GE|
with magnetization preparation
Ultrafast gradient echo sequences use a small flip angle, a very short TR and optimized k-space filling to reduce acquisition time (to roughly one second per slice). The drawback of a small flip angle and very short TR is poor T1-weighting.
To preserve T1 contrast, a 180° inversion pulse prepares magnetization before repetitions of the ultrafast gradient echo imagery sequence . Effective inversion time will correspond to the delay between the inversion pulse and acquisition of the central k-space lines.
To obtain T2-weighting, the preparatory pattern is of the spin echo type (90° - 180°) so that the ultrafast GE imaging sequence can start with longitudinal magnetization whose amplitude depends on T2 .
K-space filling can follow variable trajectories (linear, centripetal, centrifugal) to determine image contrast. Either all the k-space lines are acquired after a single inversion pulse (« single shot »), or only a group of lines is acquired (segmented filling).
The speed of these T1-weighted gradient echo sequences is particularly suited to monitoring Gadolinium bolus arrival in imaging at arterial phase and in T1 high resolution 3D imaging.
Pourquoi avoir une gestion en temps en réel de la dose d'irradiation des patients ?
Agfa HealthCare Special Report Dose Management: because life is precious.
Ziehm Vision RFD Hybrid Edition: Mobility and flexibility – the difference in an emergency
Olea Medical® literature meta-analysis on CTP thresholds in acute stroke
Olea Medical® case report on early & late MR diffusion follow-up of a stroke using CT perfusion.