Reconstruction in the image domain
This type of reconstruction is used by algorithms in the SENSE group.
Sensitivity profiles are established for each coil element whose covered spatial zone is systematically mapped. Spatial encoding in the direction of the phase encoding gradient is undersampled to save time (which is equivalent to reducing the field of view of the coil element). The intermediary images of each coil element thus present foldover artifacts. Thanks to the sensitivity profiles of the elements, the “unfolded” image can be calculated by deducing pixel value from the various intermediary images of each element. Coil element sensitivity profiles in phased array can be measured:
- either by separate acquisition before the imaging sequence (SENSE) in the form of low resolution 3D acquisition over the entire field of view
- or by self-calibration, measuring some of the missing lines in the center of the k-space (mSENSE).
Calibration is essential as it forms the basis for reconstructing the final image, and any noise in the data of an element profile will be reflected in the reconstructed image.
Olea Medical® literature meta-analysis on CTP thresholds in acute stroke
Pourquoi avoir une gestion en temps en réel de la dose d'irradiation des patients ?
BRACCO IMAGING FRANCE
Olea Medical® case report on early & late MR diffusion follow-up of a stroke using CT perfusion.
Agfa HealthCare Special Report Dose Management: because life is precious.
Ziehm Vision RFD Hybrid Edition: Mobility and flexibility – the difference in an emergency